MODEL: MT10/1 , MT10/3 , MT10/10
Pump control of relay pump stations and manhole pump stations
Relay unit with sequential logic circuit (RE9000) ignores affection from scum, suspended solids or other contamination.
Having no moving parts, the sensor offers long service life.
Mounting the sensor is easy with a cleaner and a mounting bracket.
DIP switch on relay unit allows measurement of various liquids.
(Sensitivity: 1kΩ to 80kΩ)
Principle of operation
The sensor has a pair of electrodes, and applies alternating current to them. When the liquid covers the electrodes, it forms a part of an electric circuit, causing current to flow. The sensor measures electric resistance of the liquid, and compares the measured value with a preset value to determine liquid presence. When liquid presence is detected, the sensor sends a signal to an amplifier to open or close relay contacts of relay units.
Sensor uses conductivity between the electrode and the ground to detect a level. For FRP or other plastic tanks, an earth electrode is required.